Read or Download Agilent 89410A Analyzer (installation and verification) PDF
Best nonfiction_6 books
- Experiments. Planning, analysis, and optimization
- Higher Educ. - How Colleges Are Wasting Our Money...
- Distributed Sensor Systems: Practice and Applications
- Airframe & Systems Tutorial - PPSC
- Differential Eqns - Their Soln Using Symmetries
Additional info for Agilent 89410A Analyzer (installation and verification)
For detailed illustrations of equipment setup, see the setup illustrations starting on page 2-14. If you want to pause the program and return the Agilent 89410A DC-10 MHz Vector Signal Analyzer to front panel control, press [ BASIC ]. To continue the program, press [ Display ] [ BASIC display format ] [ lower ] [ BASIC ] [ continue ]. If you changed any instrument setup states, press [ RESTART TEST ] to ensure accurate measurement results. 2-12 Agilent 89410A Verifying Specifications To run the program in manual mode To run the program in manual mode Use this procedure if you want to run the program in manual mode.
Trigger level and slope are then verified by reading the signal level and slope at 0 seconds in the time record. 1 2-26 Agilent 89410A Verifying Specifications To set up the external arm test To set up the external arm test Performance Test and Operation Verification This test verifies that the Agilent 89410A meets its trigger specification for external arm. In this test, a signal is connected to the external arm input. The signal level is increased until the instrument is armed. 1 2-27 Verifying Specifications To set up the harmonic distortion test Agilent 89410A To set up the harmonic distortion test Performance Test and Operation Verification This test verifies that the Agilent 89410A meets its dynamic range specification for harmonic distortion.
This test takes approximately one minute to complete and requires no external equipment. 1 2-14 Agilent 89410A Verifying Specifications To set up the frequency accuracy test To set up the frequency accuracy test Performance Test and Operation Verification The Agilent 89410A must be on for 30 minutes before performing this test. This test verifies that the Agilent 89410A meets its frequency accuracy specification. In this test, the analyzer measures the frequency of an accurate 9 MHz signal. The frequency limits are then calculated using the number of months since the last frequency reference adjustment.